In order to minimise the micro movement that can cause polyethylene (PE) wear at the metal shell - PE liner interface, a special locking mechanism has been developed. This allows the use of a BIOLOX® delta ceramic insert or a PE insert with the same cup design. The PE inserts are sterilised with ethylene oxide gas to prevent oxidative pre-degradation of the material. Alternatively PE inserts made of implacross® crosslinked polyethylene are available which demonstrated improved wear characteristics during preclinical tests.
EcoFit® cups have a central hole, in the acetabular base, for attaching the introducer for secure and controlled implant impaction. Once the cup is seated this central hole can be covered with a central screw plug. The EcoFit® cup also has three pre-plugged holes which can be removed, even when the cup is in-situ, for the insertion of supplementary cancellous screws to enhance primary stability (if required). These screws can be angled up to 15°. The EcoFit® NH cup is also available which doesn’t have any screw holes – only the refillable impaction hole. All versions are available with all coating options.
A 32mm articulation is possible with a 46mm diameter cup and a 36mm articulation is possible with a 50mm cup. This applies to implacross® and to BIOLOX® delta liners.
implatan®; TiAl6V4 according to ISO 5832-3
This area is intended for medical professionals only and may only be provided to certain professional personnel.